This vignette shows a few examples for the R-package ‘RLumModel’. The main function model_LuminescenceSignals()
and their arguments will be explained. All calculations were done with ‘RLumModel’ 0.2.2 and ‘Luminescence’ 0.7.5.
The output from the main function model_LuminescenceSignals()
is of class RLum.Analysis
(Kreutzer et al., 2012) and contains data of class RLum.Data.Curve
in the slot ‘records’. The advantage of this infrastructure is that the package ‘Luminescence’ offers a lot of methods to visualize and manipulate data.
All simualted data are stored in the slot ‘records’: TL/OSL/RF curves as well as the concentrations of every energy level from every step.
The following code loads a data set provided by the ‘RLumModel’ package and shows how to seperate TL/OSL/RF data from concentrations and how to visualize them.
data("ExampleData.ModelOutput", package = "RLumModel")
##show class
class(model.output)
##show structure
Luminescence::structure_RLum(model.output)
##seperate TL-curve from TL-concentrations
TL_curve <- Luminescence::get_RLum(model.output, recordType = "TL$")
TL_conc <- Luminescence::get_RLum(model.output, recordType = "(TL)", drop = FALSE)
##also possible: TL_curve <- get_RLum(model.output, record.id = 1)
##plot results
Luminescence::plot_RLum(TL_curve)
Luminescence::plot_RLum(TL_conc)
Some notes to the code example above:
drop = FALSE
was used to keep the RLum.Analysis
class for ‘TL_conc’.plot.single = TRUE
in plot_RLum()
. For more details see the manual of ‘Luminescence’.##plot every energy-level by an extra plot
Luminescence::plot_RLum(TL_conc, plot.single = TRUE)
It is also possible to choose a RLum.Data.Curve
by their ‘record.id’, which can be seen with:
##see structure of model.output
Luminescence::structure_RLum(model.output)
The first argument required for the function model_LuminescenceSignals()
is the name of a quartz luminescence model to be used, respectively the used parameter set in this quartz luminescence model. All currently implemented quartz luminescence models were described in Friedrich et al. (2016). The command to select a set of parameters from a specific model in RLumModel
is a character string with the name of the author and the year, e.g.
model <- "Bailey2001"
The available models are “Bailey2001”, “Bailey2002”, “Bailey2004”, “Pagonis2007”, “Pagonis2008”,“Friedrich2017” and “customized” (Bailey (2001), Bailey (2002), Bailey (2004), Pagonis et al. (2007), V. Pagonis et al. (2008), Friedrich et al. (2017)). For customized or own parameter sets, see vignette RLumModel - Using own parameter sets
The corresponding parameter set will be loaded automatically with the function call.
The second argument in the model_LuminescenceSignals()
function is the sequence to be simulated. There are three different ways of creating a sequence.
For all sequences, temperature differences between sequence steps are automatically simulated with a heating or cooling step in between. Also, after irradiating the sample, it is automatically kept at irradiation temperature for further 5 s to allow the system to relax prior to the next step (Bailey, 2001).
The first one is to use the popular and freely available Risø to build a personal sequence and to save it as a SEQ-file (*.seq). Files created by the Sequence Editor can be imported directly using the path of the SEQ-file. The package comes along with an example SEQ-file in the package folder in ‘extdata’. Thus, a potential sequence is created with
sequence <- system.file(
"extdata",
"example_SAR_cycle.SEQ",
package = "RLumModel")
or wherever the SEQ-file is stored. While in the Sequence Editor irradiation is commonly defined in seconds, performing the simulation requires a dose transformation to Gray. Therefore, the function model_LuminescenceSignals()
offers a special argument called lab.dose_rate
, representing the dose rate of the irradiation unit in the laboratory. By default, this dose rate is 1 Gy/s, but can be modified, e.g.,
lab.dose_rate <- 0.105
The second way of creating a sequence is by referring to a list with keywords and a certain order of code numbers or named values, which are shown in Table 1. With these keywords, it is possible to create quickly an R object of type list, which can be read by the model_LuminescenceSignals()
function.
ARGUMENTS | DESCRIPTION | SUB-ARGUMENTS |
---|---|---|
TL | Thermally stimulated luminescence | ’temp_begin’ [°C], ’temp_end’ [°C], ’heating_rate’ [°C/s] |
OSL | Optically stimulated luminescence | ’temp’ [°C], ’duration’ [s], ’optical_power’ [%] |
ILL | Illumination | ’temp’ [°C], ’duration’ [s], ’optical_power’ [%] |
LM_OSL | Linear modulated OSL | ’temp’ [°C], ’duration’ [s], optional: ’start_power’ [%], ’end_power’ [%] |
RF | Radiofluorescence | ’temp’ [°C], ’dose’ [Gy], ’dose_rate’ [Gy/s] |
RF_heating | RF during heating/cooling | ’temp_begin’ [°C], ’temp_end’ [°C], ’heating rate’ [°C/s], ‘dose_rate’ [Gy/s] |
IRR | Irradiation | ’temp’ [°C], ’dose’ [Gy], ’dose_rate’ [Gy/s] |
CH | Cutheat | ’temp’ [°C], optional: ’duration’ [s], ’heating_rate’ [°C/s] |
PH | Preheat | ’temp’ [°C], ’duration’ [s], optional: ’heating_rate’ [°C/s] |
PAUSE | Pause | ’temp’ [°C], ’duration [s]’ |
Some examples to this kind of sequence creating:
sequence <- list(
IRR = c(temp = 20, dose = 10, dose_rate = 1),
TL = c(temp_begin = 20, temp_end = 400 , heating_rate = 5))
This sequences describes an irradiation simulation at 20 °C with a dose of 10 Gy and a dose rate of 1 Gy/s, which is followed by a TL simulation from 20 °C to 400 °C with a heating rate of 5 °C/s. Note that it is important that for each sequence keyword like ‘IRR’ or ‘TL’ either the vector has to be named or the correct order of arguments is used, see ‘sub-arguments’ in Table 1. Thus the above mentioned code is equivalent to the following one:
sequence <- list(
IRR = c(20, 10, 1),
TL = c(20, 400, 5))
However, to create a SAR or dose-recovery-test (DRT) sequence with the Risø Sequence Editor or with keywords is time-consuming, because it contains a lot of individual sequence steps (preheat, optical stimulation, irradiation, …). Therefore, a third way was implemented in ‘RLumModel’ to create a (SAR) sequence after Murray and Wintle (2000) with the (required) keywords RegDose, TestDose, PH, CH and OSL temp. In addition to these keywords, the user is able to set more detailed parameters for the SAR sequence, see Table 2:
ABBREVIATION | DESCRIPTION | EXAMPLE ARGUMENTS |
---|---|---|
RegDose | Dose points of the regenerative cycles [Gy] | c(0, 80, 140, 260, 320, 0, 80) |
TestDose | Test dose for the SAR cycles [Gy] | 50 |
PH | Temperature of the preheat [°C] | 240 |
CH | Temperature of the cutheat [°C] | 200 |
OSL_temp | Temperature of OSL read out [°C] | 125 |
OSL_duration | Duration of OSL read out [s] | default: 40 |
Irr_temp | Temperature of irradiation [°C] | default: 20 |
PH_duration | Duration of the preheat [s] | default: 10 |
dose_rate | Dose rate of the laboratory irradiation source [Gy/s] | default: 1 |
optical_power | Percentage of the full illumination power [%] | default: 90 |
Irr_2recover | Dose to be recovered in a dose-recovery-test [Gy] | 20 |
So a possible DRT sequence could be the next code example:
sequence <- list(
RegDose = c(0,10,20,50,90,0,10),
TestDose = 2,
PH = 220,
CH = 220,
OSL_temp = 125,
Irr_2recover = 20)
This sequence describes a DRT, where a dose of 20 Gy will be recovered with this test. The regenerative doses are definded as 0 (natural), 10 Gy, 20 Gy, 50 Gy, 90 Gy and for recuperation and recycling ratio 0 Gy and 10 Gy, respectively. The test dose is definded as 2 Gy. Preheat and Cutheat are at 220 °C and all OSL measurements are simualted at 125 °C. There are more options to set, see Table 2.
The RLumModel function model_LuminescenceSignals()
is able to interprete this (sequence-) list as a DRT sequence.
First of all, a sequence is needed, which produces a TL signal after the sample has received a dose:
sequence <- list (
IRR = c (20 , 10 , 1) ,
TL = c (20 , 400 , 5))
Here, at a temperature of 20 °C a dose of 10 Gy was applied with a dose rate of 1 Gy/s followed by a TL measurement from 20 °C to 400 °C with a heating rate of 5 °C/s. Running this sequence with the model_LuminescenceSignals()
function produces a model output:
model.output <- model_LuminescenceSignals(
model = "Bailey2001",
sequence = sequence,
verbose = FALSE)
This results in a TL curve like the one published in (Bailey (2001), Fig. 1), see figure above. In a further step, it is easy to produce known TL phenomena like the shift of the TL peak with varying heating rate. For this purpose, a loop over a TL simulation changes the heating rate in each run.
##set heating rate
heating.rate <- seq(from = 2, to = 10, by = 2)
##model signals
##"verbose = FALSE" for no terminal output
## "TL$" for exact matching TL and not (TL)
model.output <- lapply(heating.rate, function(x){
sequence <- list(
IRR = c(20, 10, 1),
TL = c(20, 400, x))
TL_data <- model_LuminescenceSignals(
sequence = sequence,
model = "Bailey2001",
plot = FALSE,
verbose = FALSE)
return(Luminescence::get_RLum(TL_data, recordType = "TL$", drop = FALSE))
})
##merge output
model.output.merged <- merge_RLum(model.output)
##plot results
plot_RLum(
object = model.output.merged,
xlab = "Temperature [\u00B0C]",
ylab = "TL signal [a.u.]",
main = "TL signal with different heating rates",
legend.text = paste(heating.rate, "°C/s"),
combine = TRUE)
Some notes to the code above:
RLum.Analysis
object, because of drop = FALSErecordType = TL$
is necessary to match the character TL exact and not the concentrationsmerge_RLum
is necessary to merge all the single RLum.Analysis
objects in the list ‘model.output’ together to one RLum.Analysis
objectmodel = "..."
has to be changed (see Sec. 2)Another frequently simulated phenomenon is the sensitisation of quartz TL by \(\beta\)- or \(\gamma\)-irradiation followed by activation at high temperatures (Zimmerman, 1971, Pagonis et al. (2003), V Pagonis et al. (2008), Adamiec et al. (2004)), termed as thermal activation characteristics (TACs). For a simulation sequence, the reader is referred to V Pagonis et al. (2008), Tab. 1. To simulate this phenomenon with the model_LuminescenceSignals()
function, a loop causing a stepwise increase of the activation temperature is needed. The resulting intensity of the 110 °C TL peak can be plotted against the activation temperature, which shows TAC for the model parameters of “Pagonis2007”.
##set temperature
act.temp <- seq(from = 80, to = 600, by = 20)
##loop over temperature
model.output <- vapply(X = act.temp, FUN = function(x) {
##set sequence, note: sequence includes sample history
sequence <- list(
IRR = c(20, 1, 1e-11),
IRR = c(20, 10, 1),
PH = c(x, 1),
IRR = c(20, 0.1, 1),
TL = c(20, 150, 5)
)
##run simulation
temp <- model_LuminescenceSignals(
sequence = sequence,
model = "Pagonis2007",
simulate_sample_history = TRUE,
plot = FALSE,
verbose = FALSE
)
## "TL$" for exact matching TL and not (TL)
TL_curve <- Luminescence::get_RLum(temp, recordType = "TL$")
##return max value in TL curve
return(max(get_RLum(TL_curve)[,2]))
}, FUN.VALUE = 1)
The function model_LuminescenceSignals()
is also capable of simulating OSL phenomena. The next figure shows the dependency of the OSL signal on the power density of illumination for the model “Bailey2004”.
##set optical power [%]
optical_power <- c(0,20,40,60,80,100)
##loop over power
model.output <- lapply(optical_power, function(x){
##set sequence
sequence <- list(
IRR = c(20, 50, 1),
PH = c(220, 10, 5),
OSL = c(125, 50, x))
data <- model_LuminescenceSignals(
sequence = sequence,
model = "Bailey2004",
plot = FALSE,
verbose = FALSE)
##"OSL$" for exact matching OSL and not (OSL)
return(Luminescence::get_RLum(data, recordType = "OSL$", drop = FALSE))
})
##merge output
model.output.merged <- Luminescence::merge_RLum(model.output)
##plot results
Luminescence::plot_RLum(
object = model.output.merged,
xlab = "Illumination time [s]",
ylab = "OSL signal [a.u.]",
legend.text = paste("Optical power ", 20 * optical_power / 100," mW/cm^2"),
combine = TRUE
)
For simulating a DRT, it is necessary to define a sequence with the keyword Irr_2recover
, as mentioned in Section 4.3.
It should be mentioned that a simulation of a combinded PHPT and DRT may be very time-consuming, because for every preheat temperature a complete SAR cycle has to be run. A typical DRT sequence featuring various PH temperatures in ‘RLumModel’ is listed below. Note that in such a DRT simulation a loop over different preheat temperatures has to be written, utilising characteristic parameters from the literature. The test dose is set to 10% and the regeneration dose points to 40%, 70%, 130%, 160%, 0% and 40% of the recovery dose.
The data created by ‘RLumModel’ can be directly passed to the functions Luminescence::analyse_SAR.CWOSL()
and Luminescence::plot_DRTResults()
for routine analyses and plotting.
##set PH temperatures
PH_temp <- seq(from = 160, to = 300, by = 20)
##set regeneration doses
RegDose = c(0, 80, 140, 260, 320, 0, 80)
##loop over PH temperatures
DRT.output <- lapply(PH_temp, function(x){
sequence <- list(
RegDose = RegDose,
TestDose = 20,
PH = x,
CH = x,
OSL_temp = 125,
Irr_2recover = 200)
model.output <- model_LuminescenceSignals(
sequence = sequence,
model = "Pagonis2008",
plot = FALSE,
verbose = FALSE)
results <- Luminescence::analyse_SAR.CWOSL(
object = model.output,
signal.integral.min = 1,
signal.integral.max = 7,
background.integral.min = 301,
background.integral.max = 401,
fit.method = "EXP",
dose.points = RegDose,
plot = FALSE)
temp <- get_RLum(results)
out <- data.frame(
De = temp$De,
De.error = temp$De.Error)
return(out)
})
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 179.1 | D01 = 101.51
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 179.46 | D01 = 101.46
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 180.18 | D01 = 101.4
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 180.6 | D01 = 101.41
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 182.24 | D01 = 101.44
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 179.85 | D01 = 102.26
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 166.73 | D01 = 111.51
## [plot_GrowthCurve()] Fit: EXP (interpolation) | De = 159.15 | D01 = 161.2
##output as data.frame for plot_DRTResults
DRT.result <- as.data.frame(do.call(rbind, DRT.output))
##plot DRT.results
Luminescence::plot_DRTResults(
DRT.result,
preheat = PH_temp,
given.dose = 200)
In the code above, plot = FALSE
was chosen, because a single OSL plot is not necessary to analyse a SAR sequence. To calculate a D_{e} from the produced RLum.Analysis
object ‘model.output’, the function Luminescence::analyse_SAR.CWOSL()
is suitable. After specifying a number of evaluation parameters (signal and background integration interval, dose points and fit function for the dose response curve) and the analysis process, the reduced data are stored in an RLum.Results
object, here termed ‘results’ . A background integration interval from 301 to 401 translates to the signal from 30 s to 40 s, because a channel has the default width of 0.1 s. Accordingly, the signal integral ranges from 0.1 s to 0.7 s.
##set RegDose
RegDose = c(0, 80, 140, 260, 320, 0, 80)
##set sequence
sequence <- list(
RegDose = RegDose,
TestDose = 20,
PH = 220,
CH = 220,
OSL_temp = 125
)
##model
model.output <- model_LuminescenceSignals(
sequence = sequence,
model = "Pagonis2008",
plot = FALSE,
verbose = FALSE
)
##analyse SAR sequence and plot only the resulting growth curve
results <-Luminescence::analyse_SAR.CWOSL(
model.output,
signal.integral.min = 1,
signal.integral.max = 7,
background.integral.min = 301,
background.integral.max = 401,
fit.method = "EXP",
dose.points = RegDose,
verbose = FALSE,
plot.single = c(6)
)
Adamiec, G., Garcia-Talavera, M., Bailey, R.M., La Torre, P.I. de, 2004. Application of a genetic algorithm to finding parameter values for numerical simulation of quartz luminescence. Geochronometria 23, 9–14.
Bailey, R., 2004. Paper I - Simulation of dose absorption in quartz over geological timescales and its implications for the precision and accuracy of optical dating. Radiation Measurements 38, 299–310.
Bailey, R., 2002. Simulations of variability in the luminescence characteristics of natural quartz and its implications for estimates of absorbed dose. Radiation Protection Dosimetry 100, 33–38.
Bailey, R.M., 2001. Towards a general kinetic model for optically and thermally stimulated luminescence of quartz. Radiation Measurements 33, 17–45.
Friedrich, J., Kreutzer, S., Schmidt, C., 2016. Solving ordinary differential equations to understand luminescence: ’RLumModel’, an advanced research tool for simulating luminescence in quartz using R. Quaternary Geochronology 35, 88–100.
Friedrich, J., Pagonis, V., Chen, R., Kreutzer, S., Schmidt, C., 2017. Quartz radiofluorescence: A modelling approach. Journal of Luminescence 186, 318–325.
Kreutzer, S., Schmidt, C., Fuchs, M.C., Dietze, M., Fischer, M., Fuchs, M., 2012. Introducing an R package for luminescence dating analysis. Ancient TL 30, 1–8.
Pagonis, V., Balsamo, E., Barnold, C., Duling, K., McCole, S., 2008. Simulations of the predose technique for retrospective dosimetry and authenticity testing. Radiation Measurements 43, 1343–1353.
Pagonis, V., Chen, R., Wintle, A., 2007. Modelling thermal transfer in optically stimulated luminescence of quartz. Journal of Physics D: Applied Physics 40, 998.
Pagonis, V., Kitis, G., Chen, R., 2003. Applicability of the Zimmerman predose model in the thermoluminescence of predosed and annealed synthetic quartz samples. Radiation Measurements 37, 267–274.
Pagonis, V., Wintle, A., Chen, R., Wang, X., 2008. A theoretical model for a new dating protocol for quartz based on thermally transferred OSL (TT-OSL). Radiation Measurements 43, 704–708.
Zimmerman, J., 1971. The radiation-induced increase of the 100\(\,^\circ\)C thermoluminescence sensitivity of fired quartz. Journal of Physics C: Solid State Physics 4, 3265–3276.